The Aerobika* Oscillating PEP device disassembles into four parts for
cleaning. To disassemble press the tabs on either
side of the device and gently pull the top and bottom case
apart. Then remove the valve cartridge and
mouthpiece. Do not disassemble any part of the valve
cartridge. Wash the four parts in warm soapy water allowing the parts to soak for 15
minutes. Agitate gently. Rinse in warm
distilled water and shake out any excess water. Allow
parts to air dry thoroughly before you reassemble the device. The device is also dishwasher safe.
Secure the device in a basket on the top rack and run
through a light or normal cycle. Shake out any excess
water and always allow the device to air-dry
thoroughly before reassembly. To reassemble the
device slip the valve cartridge into the
bottom case. If it does not fit rotate the cartridge and try again. Reattach the top and
bottom cases until you hear the tabs click into place.
Ensure they are securely engaged by firmly pressing the two halves
together. Reattach the mouthpiece. The Aerobika*
Oscillating PEP device should be disinfected once
each treatment day using any of these methods. If you have any questions please ask your health care provider.